Pathlength calibration for the beverage module does not work in ISIscan.
CAUSE: The WinScan calibration for the beverage module is an older MLR format that does not work in ISIscan. This result is that 0.00 is predicted, or the same value is predicted always.
An updated pathlength calibration is available that will work in ISIscan.
VERSIONS AFFECTED: ALL
SOLUTION: Please click on the path below to download the updated pathlength calibration. Make a new prediction model based on this and the pathlength will be predicted properly.
See related Item: http://winisi.com/download_files/pathlen2.zip
Click here to send suggestions for FAQs to webmaster